ANDÚJAR (the anc. Sliturgi), a town of southern Spain, in the province of Jaén; on the right bank of the river Guadalquivir and the Madrid-Cordova railway. Pop. (1900) 16,302. Andújar is widely known for its porous earthenware jars, called alcarrazas, which keep water cool in the hottest weather, and are manufactured from a whitish clay found in the neighbourhood.
ANECDOTE (from ἀν-, privative, and ἐκδίδωμι, to give out or publish), a word originally meaning something not published. It has now two distinct significations. The primary one is something not published, in which sense it has been used to denote either secret histories—Procopius, e.g., gives this as one of the titles of his secret history of Justinian’s court—or portions of ancient writers which have remained long in manuscript and are edited for the first time. Of such anecdota there are many collections; the earliest was probably L. A. Muratori’s, in 1709. In the more general and popular acceptation of the word, however, anecdotes are short accounts of detached interesting particulars. Of such anecdotes the collections are almost infinite; the best in many respects is that compiled by T. Byerley (d. 1826) and J. Clinton Robertson (d. 1852), known as the Percy Anecdotes (1820–1823).
ANEL, DOMINIQUE (1679–1730), French surgeon, was born at Toulouse about 1679. After studying at Montpellier and Paris, he served as surgeon-major in the French army in Alsace; then after two years at Vienna he went to Italy and served in the Austrian army. In 1710 he was teaching surgery in Rouen, whence he went to Genoa, and in 1716 he was practising in Paris. He died about 1730. He was celebrated for his successful surgical treatment of fistula lacrymalis, and while at Genoa invented for use in connexion with the operation the fine-pointed syringe still known by his name.
ANEMOMETER (from Gr. ἄνεμος, wind, and μέτρον, a measure), an instrument for measuring either the velocity or the pressure of the wind. Anemometers may be divided into two classes, (1) those that measure the velocity, (2) those that measure the pressure of the wind, but inasmuch as there is a close connexion between the pressure and the velocity, a suitable anemometer of either class will give information about both these quantities.
Velocity anemometers may again be subdivided into two classes, (1) those which do not require a wind vane or weathercock, (2) those which do. The Robinson anemometer, invented (1846) by Dr Thomas Romney Robinson, of Armagh Observatory, is the best-known and most generally used instrument, and belongs to the first of these. It consists of four hemispherical cups, mounted one on each end of a pair of horizontal arms, which lie at right angles to each other and form a cross. A vertical axis round which the cups turn passes through the centre of the cross; a train of wheel-work counts up the number of turns which this axis makes, and from the number of turns made in any given time the velocity of the wind during that time is calculated. The cups are placed symmetrically on the end of the arms, and it is easy to see that the wind always has the hollow of one cup presented to it; the back of the cup on the opposite end of the cross also faces the wind, but the pressure on it is naturally less, and hence a continual rotation is produced; each cup in turn as it comes round providing the necessary force. The two great merits of this anemometer are its simplicity and the absence of a wind vane; on the other hand it is not well adapted to leaving a record on paper of the actual velocity at any definite instant, and hence it leaves a short but violent gust unrecorded. Unfortunately, when Dr Robinson first designed his anemometer, he stated that no matter what the size of the cups or the length of the arms, the cups always moved with one-third of the velocity of the wind. This result was apparently confirmed by some independent experiments, but it is very far from the truth, for it is now known that the actual ratio, or factor as it is commonly called, of the velocity of the wind to that of the cups depends very largely on the dimensions of the cups and arms, and may have almost any value between two and a little over three. The result has been that wind velocities published in many official publications have often been in error by nearly 50%.
The other forms of velocity anemometer may be described as belonging to the windmill type. In the Robinson anemometer the axis of rotation is vertical, but with this subdivision the axis of rotation must be parallel to the direction of the wind and therefore horizontal. Furthermore, since the wind varies in direction and the axis has to follow its changes, a wind vane or some other contrivance to fulfil the same purpose must be employed. This type of instrument is very little used in England, but seems to be more in favour in France. In cases where the direction of the air motion is always the same, as in the ventilating shafts of mines and buildings for instance, these anemometers, known, however, as air meters, are employed, and give most satisfactory results.
Anemometers which measure the pressure may be divided into the plate and tube classes, but the former term must be taken as including a good many miscellaneous forms. The simplest type of this form consists of a flat plate, which is usually square or circular, while a wind vane keeps this exposed normally to the wind, and the pressure of the wind on its face is balanced by a spring. The distortion of the spring determines the actual force which the wind is exerting on the plate, and this is either read off on a suitable gauge, or leaves a record in the ordinary way by means of a pen writing on a sheet of paper moved by clockwork. Instruments of this kind have been in use for a long series of years, and have recorded pressures up to and even exceeding 60 lb per sq. ft., but it is now fairly certain that these high values are erroneous, and due, not to the wind, but to faulty design of the anemometer.
The fact is that the wind is continually varying in force, and while the ordinary pressure plate is admirably adapted for measuring the force of a steady and uniform wind, it is entirely unsuitable for following the rapid fluctuations of the natural wind. To make matters worse, the pen which records the motion of the plate is often connected with it by an extensive system of chains and levers. A violent gust strikes the plate, which is driven back and carried by its own momentum far past the position in which a steady wind of the same force would place it; by the time the motion has reached the pen it has been greatly exaggerated by the springiness of the connexion, and not only is the plate itself driven too far back, but also its position is wrongly recorded by the pen; the combined errors act the same way, and more than double the real maximum pressure may be indicated on the chart.
A modification of the ordinary pressure-plate has recently been designed. In this arrangement a catch is provided so that the plate being once driven back by the wind cannot return until released by hand; but the catch does not prevent the plate being driven back farther by a gust stronger than the last one that moved it. Examples of these plates are erected on the west coast of England, where in the winter fierce gales often occur; a pressure of 30 lb per sq. ft. has not been shown by them, and instances exceeding 20 lb are extremely rare.
Many other modifications have been used and suggested. Probably a sphere would prove most useful for a pressure anemometer, since owing to its symmetrical shape it would not require a weathercock. A small light sphere hanging from the end of 30 or 40 ft. of fine sewing cotton has been employed to measure the wind velocity passing over a kite, the tension of the cotton being recorded, and this plan has given satisfactory results.
Lind’s anemometer, which consists simply of a U tube containing liquid with one end bent into a horizontal direction to face the wind, is perhaps the original form from which the tube class of instrument has sprung. If the wind blows into the mouth of a tube it causes an increase of pressure inside and also of course an equal increase in all closed vessels with which the mouth is in airtight communication. If it blows horizontally over the open end of a vertical tube it causes a decrease of pressure, but this fact is not of any practical use in anemometry, because the magnitude of the decrease depends on the wind striking the tube exactly at right angles to its axis, the most trifling departure from the true direction causing great variations in the magnitude. The pressure tube anemometer (fig. 1) utilizes the increased pressure in the open mouth of a straight tube facing the wind, and the decrease